Do I need to have an electrical outlet in the MR scan room for the BOLDScreen 32?
No, the power supply is meant to be powered from outside the MR scan room. A voltage converter unit (VCU) with RF filter is supplied which is to be mounted within the penetration panel, most often between the equipment room and scan room. An 11mm hole (BNC sized) is required for the mounting of the VCU.
Leave a Comment